In this paper, we proposed the selective and ultrasensitive detection of Cadmium ions using a Cysteamine functionalized microcantilever-based sensor with cross-linked DL-glyceraldehyde. The detection time for various laboratory-based techniques is generally 12-24 hours. The experiments performed to create the SAMs of Cysteamine (Cys) cross-linked with DL-glyceraldehyde on the Microcantilever surface to selectivity capture the targeted Cd2+ Ions. The proposed portable microfluidic platform able to achieve the detection in 20-23 minutes and having a Limit of Detection (LOD) of 0.56 ng (2.78 pM), which perfectly describes its excellent performance over other reported techniques. Many researchers used Nanoparticles (NPs)-Based sensors for Heavy metal ions detection, but day by day, increasing usage and commercialization of Nanoparticles are rapidly expanding their deleterious effect on human health and the environment. The proposed technique uses the blend of thin-film and microcantilever (MEMS)-technology to overdrive the disadvantages of the Nanoparticles-approaches for selective Cd2+ ions detection and having LOD less than WHO limit of 3 µg/L. The fabricated sensor can achieve the limits efficiently well below the standards set by the WHO and helpful for the early detection of HMIs polluted source.
Keywords: Piezoresistive Sensors; BioMEMS; Microfluidics; Microcantilevers; SAM (Self Assembled Monolayer's); Heavy Metal Ions (HMIs); Limit of detection (LOD); WHO (World Health Organization), Micro-ElectroMechanical Systems (MEMS)
When a peer-reviewed version of this preprint is available in the Beilstein Journals, this information will be updated in the information box above. If no peer-reviewed version is available, please cite this preprint using the following information:
Rotake, D. R.; Darji, A. D.; Kale, N. S. Beilstein Arch. 2020, 202044. doi:10.3762/bxiv.2020.44.v1
|Download RIS (Reference Manager)||Download BIB (BIBTEX)|